How to connect test leads to IP67 of DMM Aktakom AMM-1028? Find the answer in this piece of news
Aktakom AMM-1028 is a good combination of easy-to-use functions, safety and reliability. It features IP67 dust and waterproof, True RMS, rugged design for heavy duty use. The instrument complies with IEC/EN 61010-1:2001-02 and IEC/EN 61010-031:2002, Class2, double insulation.
Devices made in accordance with the requirements of IP67 (Waterproof) have tight test lead jacks. It is important that the insertion of the test leads is performed without gaps. The edges of test leads have special grooves for sealing against water.
Therefore the test leads should be inserted with a lot of effort (pushed forcefully into the jacks of the instrument) in order to achieve tight connection. Otherwise the test leads will not be securely connected to the jacks of the measuring instrument.
Tight connection of the test leads to the instrument is an essential condition for all Waterproof measurements in accordance with the terms of IP67. To verify the correct connection mode, set rotary switch to the continuity mode and press the MODE button.
Much more information like frequency asked questions and application is available on page of AMM-1028 multimeter in the self-named tabs.
Back to the list