Light measurement with zero adjustment by push button
Type K, Type J thermocouple thermometer
Real time SD memory card Datalogger, Built-in Clock and Calendar, real time data recorder, sampling time set from 1 second to 3600 seconds
Manual datalogger is available (set the sampling time to 0), during the use of manual datalogger function it can set the different position (location) No. (position 1 to position 99)
Innovation and easy operation, computer doesn’t need any extra software, after the datalogger use just take the SD card away from the meter and plug in the SD card into the computer, it can download all of the measured value with the time information (year/month/date/hour/minute/second) to the Excel directly; then users can make the further data or graphic analysis by themselves
SD card capacity: 1 GB to 16 GB
LCD with green light backlight, easy reading
Can default auto power off or manual power off
Data hold, record max. and min. reading
Microcomputer circuit, high accuracy
Power by UM3/AA ( 1.5 V ) x 6 batteries or DC 9V adapter
Function selection and measurement by AKTAKOM ATE-1537 Light meter
Turn on AKTAKOM ATE-1537 light meter by pressing Power Button.
*Pressing Power Button over 2 seconds again will turn off the meter.
The meter can select 2 kinds of functions:
- Light meter
- Type K/J Thermometer. Pressing Function Button continuously (not releasing the button) will make the LCD displaying the following text in sequence:
- LIgHT – Light meter
- tP – Type K/J Thermometer
When the Display shows the desired mode (Function) – release Function Button, the meter will execute this function on default.
Light meter function of ATE-1537 light meter
Install Probe Plug into Probe Input Socket
Power on ATE-1537 light meter by pressing Power Button. The display will show "LUX" (Ft-cd) unit. Now the meter is ready for further measuring.
Hold Sensor Handle, point the Light Sensor against the measurement light source directly. The display will show the light measurement value.
Type K/J thermometer function of ATE-1537 light meter
Select the function to Type K/J thermometer
Do not install the Humidity Probe Plug into the Probe Input Socket
Plug the Thermocouple Temp. Probe into TypeK/J Probe Input Socket. The Display will show the measuring value that sensing from the Temp. probe.
If the Display shows "K" it is ready for Type K thermometer. If the Display indicates "J" it is ready for Type J thermometer.
Preparation before datalogger function use when operating AKTAKOM ATE-1537 light meter
Insert the SD card
Prepare SD memory card (1 G to 16 G, optional), insert the SD card into SD card socket.
SD card Format
If SD card just is used for the first time with this meter, it’s recommended to make SD card format at first.
If the meter is used for the first time, you should adjust the clock time.
How can I save data from the SD card to my PC when operating AKTAKOM ATE-1537 light meter?
After Data Logger function use, take away the SD card out from the SD card socket.
Plug in the SD card into the Computer SD card slot or insert the SD card into the SD card adapter. Then connect the SD card adapter into the computer.
Power ON your PC and run EXCEL software. Download the saving data file from the SD card to the computer. The saving data will be presented in EXCEL software screen, then user can use these EXCEL data to make further Data or Graphic analysis.
AKTAKOM ATE Easy Monitor software provides the numerical indication of 4 channel measurements as well as the data recording into file in CSV format. For the device connection the use of ACE-1025/ACE-1026 USB-RS232 interface converters is required.
ADLM-A software is designed for data reading from devices of AKTAKOM ATT, ATE series, for measurement data acquisition, its processing, displaying and storage in a tablet PC with Android OS of 4.0 and more advanced versions.
ADLM-w software is designed for data reading from devices of AKTAKOM ATT, ATE series, for measurement data acquisition, its processing, displaying and storage in a PC with Windows XP x32, Windows 7 x64, Windows 7x32, Windows 8 x64, Windows Vista OS.