"Absolute measurement error" service allows you to estimate the measurement accuracy considering uncertainty (error) in the absolute value. It’s important if you take measurements at the range edge when the error value contributes to the value displayed on the device screen.
To get the value range it’d be enough to enter the estimate of the value under measurement into Value input field and to see the range of possible values taking all of the standardized errors into consideration.
Note: a point is used as a separator of the integer and fractional part.
Estimated shipping cost $12 (Continental US only!) Depending on a destination shipping cost may vary. Free shipping for orders over $450 on items weighting under 20 lbs (certain restrictions apply!) See shipping page for international rates or contact firstname.lastname@example.org
AKTAKOM AMM-3035 LCR meter (Inductance (L), Capacitance (C), and Resistance (R)) Meter features with innovative automatic selection technology, automatically measuring of R, L and C with Q, D and Θ by just turning on the meter. The testing frequency is up to 10kHz and it can output up to 100kHz test frequency. It provides high accuracy with four or five terminal Kelvin measurements.
Lower Battery Indication
Auto Power Off (10 minutes)
Sampling rate: 5times/sec
Overload Indication: "OL" or "-OL"
Auto Power Off: 10 minutes
Operating Temp 0-50°C,<=80%R.H.
Storage Temp: -20-60°C
Temp. coefficient: 0.15 (Spec.Acc)/°C, <18°C or >28°C
Safety: Comply with IEC61000 EMC standard
Power Requirement: AA size 1.5V X 4 batteries (NEDA 15A or IEC LR6)
Battery Life: 30 hours (Alkaline battery)
Special Offer 1+: Buy 1 AMM-3035 Get Second or more for $131.00 + Multi-Tip Screwdriver as a Giftcall 786 597-7752 to place an order
Attention. Discounts for various special offers are not cumulative.
In order to improve the accuracy of high/low impedance, it is recommended to do OPEN/SHORT calibration mode before measurement. Press and hold down the button for 2 seconds to enter calibration mode. The calibration procedure: OPEN ready¹ →[CAL] → OPEN calibration (30s) →[CAL] → SHORT ready² → [CAL]→ SHORT calibration (30s) . During open or short calibration processing, the 30-second countdown will be shown on LCD panels. If the calibration procedure is finished, the PASS or FAIL symbol will shown on the primary display. If PASS symbol for both OPEN and SHORT modes, the calibration data will be saved after push CAL key again.
¹OPEN ready means the input sockets or terminals have nothing connected
²SHORT ready means put a shorting bar or a short piece of conductive metal (i.e. paper clip) across the “+” and “-” input sockets or terminals.
Open Calibration (left) and Short Calibration (right)
Sorting a bunch of components
The sorting mode could help the user to make a quick sort for a bunch of components. Select the primary measurement mode (L/C/R) based on the type of components to be measured. Insert the component to be used as the “standard” reference value. Another words, insert a known “good” component that will be used for testing against all other components. Press [SORTING] button to enter to the sorting mode. The sorting mode cannot be activated unless the meter senses a component is connected to either the input sockets or terminals . When sorting mode is activated, the reference value, range and the tolerance settings can be modified.
The setting process:
[SETUP]→ range setting(use </>) → [ENTER] → reference value setting (use ∧/∨/</>) →[ENTER] → tolerance setting(use </ >) →[ENTER] →sorting mode
The tolerance range setting selection: +0.25% → ±0.5% → ±1% → ±2% → ±5% → ±10% → ±20% →+80%-20%. The default tolerance is ±1%.
In the sorting mode, the primary display to show PASS or FAIL status depends on whether the impedance measured exceeds tolerance range. The current measurement result will be shown on the secondary display. Press the [SORTING] button to exit this mode.
WARNING: If the component to be measured is a capacitor, be sure that the capacitor is fully discharged BEFORE inserting it into the input sockets or terminals. For large capacitors, it may take longer periods of time for a full discharge. Inserting a charged or partially charged capacitor into the meter’s input sockets or terminals may produce an electric hazard and may also damage the instrument, making it unusable
El medidor de AKTAKOM AMM-3035 LCR inductancia (L), capacitancia (C), y resistencia (R)). Este equipo innova con la tecnología de selección automática de R, L y C con Q, D y Θ con sólo encender el medidor. La frecuencia de las pruebas es de hasta 10 KHz y puede producir hasta 100 KHz de frecuencia de prueba. Que proporciona una gran precisión con cuatro o cinco mediciones terminales en Kelvins.