"Absolute error of my measurement" service allows evaluating measurement precision considering measurement uncertainty (or error) in absolute value. It is important in case you are making measurements at the range edge, when the value of error contributes into the value displayed by the device.
To get the range of values it’s enough to enter the evaluation of the value under measurement into Value box and to see the range of possible values including all of the standard errors.
This thermometer has a wide range of applications: food preparation, safety and fire inspectors, plastic molding, asphalt, marine and screen printing, measure ink and dryer temperature, diesel and fleet maintenance.
AKTAKOM ATE-2523 is capable of non-contact (infrared) temperature measuring at the touch of a button. The built-in laser pointer increases target accuracy while the backlight LCD and handy push-buttons combine for convenient, ergonomic operation.
Precise non-contact measurements
Built-in laser pointer
Automatic selection range and Resolution to 0.1ºC or 1ºC
ºC/°F switchable button
Automatic Data Hold & Auto power off
The meter at 10 inches away measure 1in target
Backlite LCD display
MAX/MIN temperature measurement Function
Overrange indication and automatic power off
Unique flat surface, mini modern housing design
Accuracy is given at 18°C to 28°C (64°F to 82°F), less than 80%RH
50.0°C~-20.0°C (-58.0°F~-4.0°F): ±5°C/±9°F
-20.0°C~200.0°C (-4.0°F~200.0°F): ±2% of reading or ±2°C/±4°F
200.0°C~760°C (-200°F~1400°F): ±2.5% of reading or ±2°C/±4°F
Special Offer 1+: Buy 1 ATE-2523 Get Second or more for $71.00+ Multi-Tip Screwdriver as a Giftcall 786 597-7752 to place an order
Attention. Discounts for various special offers are not cumulative.
Power supply: 9V battery, NEDA 1604A or IEC 6LR61, or equivalent
1. 9V battery
2. Carrying case
Hold the meter by its Handle Grip and point it toward the surface to be measured
Pull and hold the Trigger to turn the meter on and begin testing. The display will light if the battery is good. Replace the battery if the display does not light
While measuring, the SCAN display icon will flicker in the upper left hand corner of the LCD
While continuing to pull the Trigger:
Push the Laser button to turn on the laser pointer. When the laser is on the laser icon will appear on the LCD over the temperature. Aim the red beam approximately a half inch above the point of test (pressing the Laser button again turns the laser off).
Select the temperature units (°C or °F) using the °C/°F buttons.
Push the Backlight key to turn on the LCD backlighting function.
Push the mode key to enter the max temperature or min temperature measurement
Release the Trigger and the HOLD display icon will appear on the LCD indicating that the reading is being held
The meter will automatically power down after approximately 7 seconds after the trigger is released.